Product code: KIMPELLER - 0801
All Kestrels feature a patented user-replaceable impeller assembly.
• User replaceable
• High precision axle and low-friction Zytel® bearings
• Light weight impeller
• 25mm (1") diameter
• Factory calibrated
• Fits all Kestrel models
• Quick and easy to replace - no tools required
For customers requiring independent calibration certificates for their Kestrel, a new impeller would require re-calibration - please refer to our calibration services.
24 HOUR SHIPPINGProcessing Mon-Fri
on all in-stock models
FREE UK DELIVERYFree standard delivery
Next day delivery available
RETURN POLICYNot 100% satisfied?
Return within 21 days
NOTE: Replacement Impeller is compatible with ALL Kestrels models.
The impeller assembly is sold as a replacement impeller for all models in the Kestrel hand-held instrument range. It can also be used in other flow measurement applications.
The light weight impeller is mounted on a precision axle with low-friction Zytel® bearings to provide accurate flow measurements even at low wind speeds. A magnet is mounted on the axle so the impeller's rotation can be easily sensed.
The impeller housing is designed so no adjustment of the bearings is required and during manufacture each assembly is individually tested in a wind tunnel. This ensures that when a new impeller assembly is installed in a Kestrel instrument the accuracy of the instrument is restored to within specification.
The assembly is supplied with a retaining o-ring.
|Replacing The Impeller|
Step 1. Using your thumbs press FIRMLY on the impeller module to remove the entire unit.
Step 2. Press only the sides of the new impeller when installing to avoid damaging the precision hub bearing.
Step 3. Insert the impeller so that the side that has a small triangle (close to the perimeter) faces the front of the Kestrel when installed.
Step 4. Orient one "arm" straight up (fig.1 & fig.2).
Step 5. The impeller can be pushed in from either side.
• Replacing an Impeller on a Kestrel 1000 to Kestrel 3500 Weather Meter
• Replacing an Impeller on a Kestrel 4000 to Kestrel 4600 Weather Tracker
|Operational range||0.6m/s to 60m/s|
|Specification range||0.6m/s to 40m/s|
|Calibration drift||1% after 100hrs operation at 7m/s|
|Wind speed measurements are tested during manufacture.
(US Patent 5,783,753)
|Kestrel Impeller Assembly data sheet|
|Instructions on how to replace the impeller in your Kestrel|
|Kestrel Product Catalogue|
|Kestrel Pocket Weather Meters Product Catalogue 2012 (7MB)|